AGILENT 34970A USB DRIVER
Here is a link to the connector I am using, which I believe should be right. In fact I just hooked up a on my desk here using a “generic” USB to Serial adaptor and one of the Agilent null modem cables and tried a quick program to check. RS Communication with Agilent A. No matter how small the script, or how long, it always gives the same error on the last line. BenchVue gives you an intuitive interface for test configuration and real-time data display and analysis. Thank you so much for your continued help and patience on this!
|Date Added:||5 August 2010|
|File Size:||58.60 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Choose from eight optional plug-in modules to create a compact data logger, full-featured data acquisition system or low-cost switching unit.
Has anybody ever encountered this sort of error before? See search results instead: I really don’t know what else to recomend It is a null modem cable.
No matter how small the script, or how long, it always gives the same error on the last line. Message 1 of If we cannot see “what you did wrong”, we sgilent unable to tell you how to fix it.
Prices shown are exclusive of taxes. It seems to work fine for me Each channel can be configured independently to measure one of 11 different functions without the added cost or hassles of signal-conditioning accessories.
Both settings on the DAQ and in NI MAX match up, and I checked the interface but shorting he interface pin 2 to 3 to check if it’s reading wrong, so that’s working correctly. Also the serial cable that comes with the A is NOT a “proper cable” as it does not have the hand shaking lines only Rx, Tx, and ground.
How to Buy or Rent.
34970A Data Acquisition / Data Logger Switch Unit
Contact an Expert Chat Live. Update on the cable: Here is a link to the connector I am using, which I believe should be right.
Solved: RS Communication with Agilent A – Discussion Forums – National Instruments
I have double checked to make sure that all of the settings on the instrument match usbb settings through Keysights app, and they are both in sync.
Message 3 of I’m not sure exactly what happened that manually throwing SCPI commands at it wasn’t working out, but it’s working now, so I guess I’m on to debugging the program now. Message 10 of Counterfeits of this or closely related products have been found in the market. Bob Schor 28 August Eventually it generates an usbb on the instrument, which the guidance uwb the manual says ” Invalid Separator” even if I’m putting in the command SYST: Setup and make measurements quickly, export data or log and view real-time measurements with a variety of display types to interpret your data faster.
Message 6 of None of them seem to fix the issue.
Agilent A and Windows 10 | Keysight Community
Accepted by topic author AnthonyMN. Plug-in Module Product Comparison. Our most popular module A features a built-in thermocouple reference and 20 two-wire channels.