AFG3021B LABVIEW DRIVER DOWNLOAD
You have to wire the functions sequentially. Hi Michael, Thanks for your reply. As it states in the KB its a good idea changing the default timeout period to more than ms. Unmatched performance, versatility, intuitive operation, and affordability make the AFG Series of Function, Arbitrary Waveform, and Pulse Generators the most useful instruments in the industry. Please note True constant on the Reset input to tkafg3k Initialise vi which. Share or Embed Document. I have a few more questions
|Date Added:||13 May 2013|
|File Size:||19.19 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Anyone know how I can get around this? Use this form to quickly and easily send an email to a dealer. I have found an interesting KB that explains the error and how to get around it.
Remind me later Go.
The driver seems to come in two parts. Is this the latest version of the driver?
Tektronix AFGB Function Generator – Instrument Driver – National Instruments
Familiarize yourself with the printed operation manuals for the TDSB oscilloscope. Issuu is a digital publishing platform that makes it simple to publish magazines, catalogs, newspapers, books, and more online.
My last problem is that there seems to be no way to set the phase of a waveform; as I afg301b, I want two Sine waves 90 deg out of phase.
Most Active Hardware Boards: Users can choose from 12 different standard waveforms. Tektronix tkafg3k Signal Generator.
Afg3021b labview download
Version download access trial free Familiarize yourself with the printed operation manuals for the TDSB oscilloscope. At some point, you should address the errors that you get.
Based on your description of overall system, I think that you’re on the right track. Download Driver and Related Software Use the chart below to download the right driver for your. Send to a Friend.
On pulse waveforms, leading and trailing edge time can be set independently. De dados que fornece o valor da.
I have a few more questions Please note True constant on the Reset input to tkafg3k Initialise vi which. I want to drive the two channels with Sine waves at Hz, but 90 deg out of phase with each other can’t find a way to set the phase but that’s another problem! Send a message to this seller. I recommend going through the KB. This makes it a good companion to your TDS OpenChoice oscilloscope with its own data acquisition and analysis abilities.
Labvies of Instrument Drivers An instrument driver is a set of software routines that control a.
If I change it back to a True constant I get: Any further help much appreciated, as afg30211b is taking far too long and it should be easy! VISA resource name is correct. That’s the way the driver was intended to be used.
Provided and all seems to be well.
Tektronix AFGB Used and New Function Generators –
My last problem is that there seems to be no agf3021b to set the phase of a waveform; as I said, I want two Sine waves. External signals can be connected and added to the output signal. Dual channel models can generate two identical or completely different signals. Your password must be at least 8 characters long and contain at least one uppercase letter.